- About ElZain
- Test items
- Customer support
- Media center
THE GENERATION DIAGNOSTIC SYSTEM
ichroma™ TRIAS Reader
HOW DOES IT WORK?
Ichroma™ TRIAS Reader is an in vitro diagnostic (IVD) device that measures the concentration of targeted analyte from a sample such as blood, nasal swabs, and nasopharyngeal swab in a quantitative or semi-quantitative way. The ichroma™ TRIAS Reader is a d
|Interface||Expansion of user interface through adoption of Android OS|
|Display||10.1” TFT "touch panel provides operational stability and quick response|
|Data storage and update||Available to store and search large amount of test results - Patient test results (5,000), Quality control (5,000), ID Chip (100), User ID (100) Data back-up and OS update using SD card|
|Optical technology||Fluorescent optical module for high sensitive / semi-quantitative test available|
|Test mode||Automatic test mode, scan mode, Single mode test, QC (Quality Control), Calibration and review (results saving) mode|
|Stability||Reduced risk of test error based on the equation saving function|
|Connectivity||Built-in printer, improved wired / wireless connectivity, LIS / HIS communication USB hub chip supporting 4 USB ports|
|Convenience||Keyboard adaptable for non-touchpad user Improved convenience to input sample ID and tester ID|
|Dimensions||210 mm (L) x 116 mm (W) x 79.8 mm (H)|
|Power supply||AC 110-240 V, 47-63 Hz, 1.1-2.2 A|
|Data output||10.1” TFT touch color LCD|
|Communication ports||USB 4 ports, LAN port, RS-232|
|Temperature range for operation||15 ~ 35 °C|
|Temperature/humidity range for storage||Temperature -20 ~ 50°C, Humidity Max. 70%|
|Data output||LCD, built-in printer, RS-232 port|
|Data storage||Patient test results (5,000) / System check results (100) Standard material (control) test results (5,000)|
|Save of ID chip information||500 ID chips (Max. 5 LOTs per each item)|
|Throughput||Max. 80 tests per hour (different depending on item)|
QThere are two modes in test mode, automatic test and scan modes. I want to know the difference.
In the automatic test mode, sample reaction occurs in the reader after dropping a sample into the cartridge and immediately inserting the cartridge into the reader. It can be automatically measured according to the set time and early judgment (1 minute 30 seconds for influenza) is capable. This automatic test mode is advantageous when patients number is small. In the scan mode, after dropping a sample into the cartridge, reacting for a pre-treatment time (10 minutes for influenza) from the outside of the reader and inserting the cartridge into the reader, the result is immediately determined by the reader. It is advantageous for large number of patients.
QOur hospital manages patients through barcode and in-hospital computer network, and the results are sent to the doctor in the diagnostic laboratory.
Can the TRIAS reader send the patient's barcode and results directly to the in-hospital network?
TRIAS can input patient ID using barcode reader and patient’s barcode is printed on output TRIAS can input patient ID using barcode reader and patient’s barcode is printed on output paper. The TRIAS reader has a LIS system thus, patient test results from the laboratory can be transmitted to the relevant department or doctor through the in-hospital computer network. The TRIAS reader has a LIS system thus, patient test results from the laboratory can be transmitted to the relevant department or doctor through the in-hospital computer network.
QIs the TRIAS reader only an influenza test?
No. The TRIAS reader is able to measure various diseases such as respiratory diseases. Currently, only influenza is used however, other respiratory diseases are under development, and they are pected to be launched soon.