- About ElZain
- Test items
- Customer support
- Media center
THE GENERATION DIAGNOSTIC SYSTEM
HOW DOES IT WORK?
The i-chamber is an auxiliary device for the ichroma (ichroma ™, ichroma ™ II) systems and provides an environment set to the correct temperature to the iChroma cartridge when heating is required between tests. You can improve the reliability of your test
|Interface||Expansion of user interface through graphic interface|
|Display||3.2” TFT Color LCD providing high visibility|
|Data storage and update||Available to store and search large amount of test results (up to 1,000) Test results are easily distinguished from calibrator measurement results Convenient update using ID chip|
|optical technology||Absorbance analysis system|
|Connectivity||Data output available through external printer connection|
|Convenience||Finger-prick and tube blood available (Finger blood collection and pipetting available)|
|Dimensions||190 mm (L) x 120.6 mm (W) x 108 mm (H)|
|Weight||1.5 kg (± 0.3 kg)|
|Number of cartridge slots||6 (Temperature can be set independently for each slot)|
|Power supply||DC 12 V, 5 A|
|Data output||LCD (16 x 4 characters)|
|Communication ports||Mini USB port (for software upgrade)|
|Environmental conditions for the operation||Temperature and humidity : 15~35°C(59~95oF), Humidity up to 70%, Installation : Dry, clean, flat and horizontal surface in an indoor setting away from direct sunlight, mechanical vibration and electromagnetic field, etc.|
|Heating/Cooling time||- About 10 minutes, when the target temperature (Td) is within 5°C of the ambient temperature - About 1 hour, if the target temperature (Td) is lower than the ambient temperature by 10°C or more|
QCan I set the temperature of the i-Chamber?
Yes. Press the two buttons simultaneously to set the target temperature, then press the two buttons again to complete the temperature setting.
QCan I start testing though temperature does not reach the target?
No. please turn on the i-Chamber and wait until the target temperature is reached. Test should be carried out when the i-Chamber provides a constant temperature condition.
QCan I set the reaction time of the i-Chamber?
Yes. Press the right button for 3 seconds, set the reaction time, and press the two buttons simultaneously to complete the time setting.